Find great deals for HP a Spectrum Analyzer Display Mainframe W/ HP b Pattern Generator. Shop with confidence on eBay!. Find great deals for Hewlett Packard HP a Display With b Optical Spectrum Analyzer. Shop with confidence on eBay!. Software and documentation supplied. This service guide is part of an Option OB3 package which includes: HP A Service Guide. HP A Component.
You really need to know what you are doing before probing and testing a thing like this live. This paid for itself in just this repair. There is a current sense resistor and 7000a is what current looked at turn-on try:.
And a final picture of the twin transformers in the working power supply.
This hhp one required some improvisation because it is a quad-transistor instead of dual, and the DIP package is very different. Many suggested changing every capacitor in the board, something I was not going to do, I wanted to find what the fault really was all capacitors turned to be ok:.
High voltage, big and the troubleshooting will need to be done live.
You’ll probably want to use an isolation transformer and differential probes, and keep one hand on your pocket at all times. I used a thin teflon tube and a copper wire to recreate the four turns that form the 70004 and guess what, it worked!!
Knowing I had nothing to loose, I went and opened the transformer can with a Dremel, and removed the resin with hot air and tweezers, carefully to not damage the windings.
The supply operates a high switching frequency and many parts of the circuit are shorted at DC, so I could not find this until I took both transformers out. After many hours of testing and probing, and and few components that exploded in the process most times my fault I found there was a transformer with a 14 ohm leak between primary and secondary.
Showing Results for Keysight Technologies (Agilent HP) 70004A
Can you tell one from the other? I built this “adapter” thing as a temporary solution, you too know what that means: Something was drawing more current than expected and the current-limit protection was tripping. I was happy to find it actually worked I confessed before that I killed 7004a few components on the process, and I could not find replacements for some of them: This is the beast on the bench, that PCB looks very timidating to me. This board is a latter revision than the ones that appear on the schematics but there are not many changes.
No hl, the secondary jacket was burned and touching the primary in one spot:.
70004A Color System Display
One day, it refused to turn on. You can see the power supply slowly increasing the current at turn on until it gets too high before it starts regulating the voltage, up quits. The block diagram was invaluable, this switching power supplies are complex. I did not know where to start.
Home Projects Music Photo. There is no way to check one of the blocks of the power supply alone, you have to give it power and hope for the best. Could this be fixed?
Keysight Technologies (Agilent HP) A used or new for sale at used-line
Do you 7004a that suspicious potting that squeezed outside the can in one of them?:. I could see the power LED flash briefly upon turn-on and then I bought this cheap A color display as faulty. I was lucky to have bought a FLUKE D differential voltage probe this year at Friedrichshafen to the nice guy with grey hair and glasses that sells mostly Fluke stuff.
Privacy & Cookies Policy
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.